Original Pipet-Aid® Pipette Controller
- Available with or without EZ Grip TC filter nosepiece
- Available with dual pump filtration unit for use with biohazardous materials
- Can be used with glass or plastic pipets
- Autoclavable components
The Drummond Pipet-Aid set the standard for a convenient alternative to mouth pipetting in the laboratory. A patented, easy-to-use volume control mechanism permits precise fluid intake and delivery.
A check valve protects against accidental fluid intake into the handle and optional Tissue Culture Nosepiece will prevent cross-contamination due to over-pipetting. A special filtration unit for use with biohazardous materials is also available.
|BUY NOW||4-000-110||Pipet-Aid, Plain, 110V|
|4-000-111||Pipet-Aid, with Dual Pump Filtration Unit, 110V|
|BUY NOW||4-000-110-TC||Pipet-Aid complete with TC Nosepiece and four extra Filters, 110V|
|BUY NOW||4-000-111-TC||Same as above but with Dual Pump Filtration Unit, 110V|
|4-000-220||Pipet-Aid, Plain, 220V, CE, Euro|
|4-000-221||Pipet-Aid, with Dual Pump Filtration Unit, 220V, CE, Euro|
|4-000-220-TC||Pipet-Aid complete with TC Nosepiece and four extra Filters, 220V, CE, Euro|
|4-000-221-TC||Same as above but with Dual Pump Filtration Unit, 220V, CE, Euro|
U.S. Patent Number: 4,624,147
|4-000-001||(1)||Standard Nosepiece, Complete|
|BUY NOW||4-000-002||(2)||Rubber Insert (2/pk)|
|BUY NOW||4-000-003||(3)||Check Valve Assembly|
|4-000-005||Filter Adapter Kit|
|4-000-006||Complete Handle Assembly|
|4-000-007||Complete Pump 110V|
|4-000-008||Internal Parts Kit with Hose|
|4-000-009||(5)||Pipette Holder Outer Shell only|
|4-000-010||Diaphragm Kit for Pump|
|4-000-011||Hose Connectors (pair)|
|BUY NOW||4-000-012-W||(6)||Valve Body only|
|4-000-014||Complete Pump 220V|
|4-000-015||Vacuum Valve Housing for Pump|
|4-000-016||Pressure Valve Housing for Pump|
|4-000-017||(8)||Handle Shell only|
|BUY NOW||4-000-019||(10)||Button Assembly with springs (pair)|
|BUY NOW||4-000-050||EZ Grip TC Nosepiece Complete Kit|
|BUY NOW||4-000-051||Self-locking TC Filter 5/pk|
|BUY NOW||4-000-051-S||Sterile Self-locking TC Filter 5/pk|
|BUY NOW||4-000-052||Self-locking TC Filter 50/pk|
|BUY NOW||4-000-052-S||Sterile Self-locking TC Filter 50/pk|
|4-000-053-A||EZ Grip TC Nosepiece Outer Shell only|
|4-000-060||Dual Pump Filters (pair)|
|BUY NOW||4-000-089||Pipet-Aid Holster-Type Wall Bracket|
All standard units operate from A.C. current. Plug the unit into an appropriate wall outlet. When ready to use, turn "on" the on/off switch to activate the pump. You will hear the pump operating. Place a pipette into the nosepiece of the handle unit by gently pushing the pipette and twisting simultaneously until a firm grip is observed.
To aspirate: Gently push the upper button. The further the button is pushed, the faster liquid will be aspirated.
To dispense: Gently push the lower button. Again, the further the button is pushed, the faster liquid will be dispensed. Caution, most standard units will have a "check valve"(white or gray), in the nosepiece to prevent drawing fluid into the handle in the event of over-pipetting. A small float inside the check valve will build up protein and occasionally stick, either open or closed. The valve can be opened up and cleaned with deionized water, dried, and reassembled. Remember, the black o-ring always goes up into the handle. Improper installation will prevent the unit from operating.
Units with T.C. nosepieces contain a self-locking filter. By design, this filter will prevent any aspiration or dispensing once contact with aqueous solutions has occurred. REPLACEMENT IS NECESSARY. Caution, make sure the rubber insert is dry and free of any liquid before using the Pipet-Aid again. This will prevent contamination of the new filter and of your next sample.
Any questions regarding the operation of your Pipet-Aid should be directed to our "HOTLINE" at 1-800-523-7480 or 610-353-0200.