Portable Pipet-Aid® XP Pipette Controller
- The lightweight, compact Portable Pipet-Aid®
- Three-speed settings provide more precise control for use in the pipet filler mode and the pipette dispenser mode
- Easy access battery replacement
- EZ Grip nosepiece with double membrane filter
- Removable stand
- Holster-type wall bracket included
Specifically designed to minimize operator fatigue, the Portable Pipet-Aid XP is smaller and lightweight. It accepts most pipettes from 1 to 100 mL.
Each pipette filler and pipet dispenser button has three speeds, (F) Fast, (M) Medium, (S) Slow. The medium speed is suitable for most routine techniques. Fast speed can deliver up to 50 mL in 5 seconds and can be used for rapid filling of large pipets or breaking up cell clumps. Slow speed is ideal when using small pipettes or when more precise delivery is required. A gravity drain model is available which substitutes a gravity drain (G) function for the fast (F) mode on the dispense button.
The XP is powered by a 9 volt (Ni-MH) rechargeable battery.
A Tissue Culture Nosepiece, which features a unique filter element, will cease to allow air to pass in either direction if it becomes wet, alerting the user to possible contamination of the rubber insert.
Weight: 7.5 oz.
Capacity: Accepts 1 to 100 mL pipets
Battery: 9 Volt Ni-MH
|BUY NOW||4-000-101||Portable Pipet-Aid XP supplied complete with 110V recharger plus four replacement filters|
|4-000-201||Same as above but 230V, CE, Euro|
|4-000-201-E||Same as above but 240V, CE, UK|
|BUY NOW||4-000-102||Portable Pipet-Aid XP with Gravity Drain, 110V|
|4-000-202||Portable Pipet-Aid XP with Gravity Drain, 230V, CE, Euro|
|4-000-202-E||Portable Pipet-Aid XP with Gravity Drain, 240V, CE, UK|
U.S. Patent Numbers: 5,214,968, 5,616,871, 5,294,405
|BUY NOW||4-000-002||(2)||Rubber Insert (2/pk)|
|BUY NOW||4-000-030||Pipet-Aid Charger 110V, (8.7 V/8 mA)|
|4-000-031||Pipet-Aid Charger 230V, Euro, (8.7 V/8 mA)|
|4-000-033||Pipet-Aid Charger 240V, UK, (8.7 V/8 mA)|
|BUY NOW||4-000-035||(5)||Ni-MH Rechargeable Battery|
|BUY NOW||4-000-040||(6)||Portable Motor and Pump|
|BUY NOW||4-000-051||(3)||Self-locking TC Filter 5/pk|
|BUY NOW||4-000-051-S||(3)||Sterile Self-locking TC Filter 5/pk|
|BUY NOW||4-000-052||Self-locking TC Filter 50/pk|
|BUY NOW||4-000-052-S||Sterile Self-locking TC Filter 50/pk|
|BUY NOW||4-000-057||(4)||TC Rubber Gaskets (pair)|
|BUY NOW||4-000-070||(14)||Button Assembly XP/XL includes springs (pair)|
|4-000-072||(7)||XP Switch Operator|
|BUY NOW||4-000-073||(8)||Valve Body XP/XL (complete with buttons, springs, and gaskets)|
|4-000-073-G||XP Valve Assembly/Gravity/with Buttons|
|4-000-074||(9)||XP Nosepiece Release|
|4-000-075||(10)||XP Handle Connector|
|4-000-076||(11)||XP Wire Harness with Microswitch and Board|
|4-000-077||(12)||XP Handle Shell includes 2 Handle Halves and 1 Battery Door|
|4-000-078||(13)||XP Battery Door|
|BUY NOW||4-000-079||EZ Grip Nosepiece (Black)|
|BUY NOW||4-000-079-B||(1)||EZ Grip Nosepiece - Blue|
|BUY NOW||4-000-079-G||(1)||EZ Grip Nosepiece - Green|
|BUY NOW||4-000-079-O||(1)||EZ Grip Nosepiece - Orange|
|BUY NOW||4-000-079-R||(1)||EZ Grip Nosepiece - Red|
|BUY NOW||4-000-079-V||(1)||EZ Grip Nosepiece - Violet|
|BUY NOW||4-000-079-Y||(1)||EZ Grip Nosepiece - Yellow|
|BUY NOW||4-000-079-Z||(1)||EZ Grip Nosepiece - Assorted|
|BUY NOW||4-000-079-W||E-Z Grip TC Housing Elite White|
|BUY NOW||4-000-087||Tissue Culture Nosepiece Kit for XP, XL, and Hood Mate (includes 1, 2, 3, 4)|
|BUY NOW||4-000-087-G||Tissue Culture Nosepiece Kit, Gray, for XP, XL, and Hood Mate (includes 1, 2, 3, 4)|
|BUY NOW||4-000-089||Pipet-Aid Holster-Type Wall Bracket|
|BUY NOW||4-000-090||Stand for XP Portable Pipet-Aid|
Speed Controls: The XP has three settings on each button to set the desired speed range (S=slow, M=medium, F=fast). The fill and empty buttons are independent of each other and can be set at different speeds. The speeds are changed by simply rotating the button until a detent is felt and the letter is upright when held in the right hand. The buttons are also proportional in nature. As with all other Pipet-Aids, the further you depress the button, faster filling or dispensing will result. With both buttons set at "M", the XP will operate at the same speed as the Portable Pipet-Aid. Rotating the button to "S" will reduce the speed to approximately half of the medium speed while rotating to "F" will double the medium speed. DO NOT CHANGE SPEED RANGES WITH LIQUID IN THE PIPETTE.
Changing the Filter & Nosepiece Removal (View a video on the "Maintenance Video" tab): Unlike other Pipet-Aids with the T.C. nosepiece, the XP has a one piece nosepiece that contains the filter and rubber insert. This nosepiece locks into place and will not loosen when removing the pipette. To remove nosepiece, slide the nosepiece release button towards the handle and turn the nosepiece one quarter turn counter-clockwise.
Battery Replacement (View a video on the "Maintenance Video" tab): The battery is contained in the upper portion of the handle. Removal of one phillips screw (B), will enable you to remove the battery compartment door which will give you access to the battery. Use only Drummond replacement batteries (4-000-035). Under normal conditions, these batteries should last approximately 2 years.
Charging the Battery: The Drummond replacement battery is a Ni-MH battery which should be charged overnight following use. No "memory" will be developed and overcharging is not a problem.
Use of the Removable Stand: This lightweight stand slides on to the top of the XP enabling the user to set the Pipet-Aid XP down without removing or emptying the pipette. The stand will prevent liquid from running back into the nosepiece and prevent the pipette tip from contacting anything thus maintaining sterility. When not needed, the stand is easily removed.
Removing the Battery
- Remove stand
- Remove screw at top right
- Flip up and remove battery door
- Disconnect battery
Nosepiece Filter Replacement
- Turn over Pipet-Aid® XP.
- Slide nosepiece release away from nosepiece while turning nosepiece counter-clockwise 1/4 turn. Remove.
- Remove rubber insert.
- Remove filter and TC gasket and discard filter only. Reuse TC gaskets.
Replacing the Battery
- Connect new battery and slide into place
- Reinstall battery door
- Replace screw
Nosepiece Filter Replacement
Replacing the Filter
- Place one TC gasket on each end of new filter. Place new filter into handle. Orient such that long stem is directed away from handle.
- Push rubber insert onto filter end.
- Reconnect nosepiece housing by pressing and turning clockwise until locked.