- Dual graduations for non-contact delivery
- With both standard length and short length plungers
The Wiretrol II provides the accuracy and convenience of the Wiretrol I but with dual graduations and a shorter plunger for noncontact delivery.
|Cat. No.||Sizes (μL)||Color Code||Pack|
|5-000-2005||1 to 5*||White||250|
|5-000-2010||5 and 10*||White||250|
|5-000-2020||10 and 20*||Orange||250|
|5-000-2050||25 and 50**||Double White||250|
|5-000-2100||50 and 100**||Green||250|
|5-000-2200||100 and 200**||Blue||250|
*Wire plunger only
**Wire Plunger with PTFE tip
|5-000-2000-L||SS Plunger, 1-5/5 & 10 μL/WRTL II, Long||1 ea.|
|5-000-2000-S||SS Plunger, 1-5/5 & 10 μL/WRTL II, Short||1 ea.|
|5-000-2001-L||SS Plunger, 10 & 20 μL/WRTL II, Long||1 ea.|
|5-000-2001-S||SS Plunger, 10 & 20 μL/WRTL II, Short||1 ea.|
|5-000-2002-L||SS Plunger, 25 & 50 μL/WRTL II, Long||1 ea.|
|5-000-2002-S||SS Plunger, 25 & 50 μL/WRTL II, Short||1 ea.|
|5-000-2003-L||SS Plunger, 50 & 100 μL/WRTL II, Long||1 ea.|
|5-000-2003-S||SS Plunger, 50 & 100 μL/WRTL II, Short||1 ea.|
|5-000-2004-L||SS Plunger, 100 & 200 μL/WRTL II, Long||1 ea.|
|5-000-2004-S||SS Plunger, 100 & 200 μL/WRTL II, Short||1 ea.|
Assembling piston to bore: insert piston into the color coded (wide printed band near one end) end of the glass bore. This end has been flame polished to prevent damage to the PTFE piston.
Insert the Wiretrol into the fluid to be dispensed. Hold the bore between the thumb and third finger and operate the plunger with the index finger. Draw up thumb and third finger and operate the plunger with the index finger. Draw up the sample until the bottom of the piston intersects the desired calibration line.
Dispense the sample by depressing the piston until the handle touches the bore. If dispensing into a diluent, simply swirling the core and tip in the fluid will remowe all of the sample.
Accuracy: ±1% T. D.
The short plunger is used where cross contamination is a problem. The plunger is aligned with the upper calibration mark and the sample is then drawn up to the lower calibration mark, one-half of the stated volume. When the plunger is depressed the sample is delivered. since the plunger has not contaminated the sample it may be reused.
Should you have any questions or wish to reorder please call Drummond at 1-800-523-7480 or 610-353-0200.